Accessories for the automatic blood sedimentation systems Sediplus® S 100 and S 2000, the manual sedimentation racks for ESR determination and the special mixers M200 and M2000 supplement the extensive blood sedimentation product range.
Yes, the Sediplus® S 2000 NX can be integrated through an interface.
No, each measuring station can be filled individually / independently of time.
Yes, the ESR can be determined automatically with the Sediplus® S 2000 NX (40 measuring stations).
For blood sedimentation, the S-Sedivette® should always be carefully mixed after blood collection. According to the CLSI standard, the measurement must be started no later than 4 hours after blood collection. The first reading of the blood sedimentation rate (ESR) should take place after 60 minutes.
The S-Sedivette® is prepared with trisodium citrate solution, which is 20% of the nominal volume. This ensures the important ratio for blood sedimentation of 1 part citrate + 4 parts blood.
The S-Sedivette® has a shelf life of 15 months.
Yes, the S-Sedivette® must be mixed thoroughly immediately after blood collection: invert the S-Sedivette® slowly five times. The air bubble / glass bead must always travel to the other end. Do not shake it back and forth quickly. For safety and to save time, we recommend the special Sarmix® M 2000 mixers.
The S-Sedivette® was developed and evaluated for determining the blood sedimentation rate (ESR) according to the Westergren method.